How should someone approach salt testing procedures?
Cat. No. 4210
Measurement Method: Conductivity method
Measurement Range: 0.0 to 10.0% (100g) of salt concentration
Minimum indication: 0.01% for salt concentration of 0.00 to 2.99% 0.1% for salt concentration of 3.0 to 10.0%
Measurement Accuracy: Displayed value +- 0.05% (for salt concentration of 0.00 to 1.00%) Relative precision +- less than 5% (for salt concentration of 1.00 to 10.0%)
Temperature Compensation: 10 to 40 degrees C (automatic temperature compensation)
Ambient Temperature: 10 to 40 degrees C
Optional: 2.84% NaCl Solution AB284 : RE-120284
Power Supply: 006P Dry battery (9V)
Dimensions & Weight: 17 x 9 x 4 cm, 300g (main unit)
This product has no reviews
You can improve your results by diluting your sample! For those samples with more salt content simply make a mixture with 10% sample and 90%water (for example) = 100%. Mix your solution and then take your reading. Since the sample only accounts for 10% of the solution the operator can then take a reading and calculate the result as a % or decimal point of the reading.